- Technical Drawings
|Water Content (%)||38|
|Dk (35°C, Fatt Units)||9|
|Refractive Index Dry||1.520|
|Refractive Index Hydrated (35°C)||1.442|
|Linear Expansion (mm)||1.205|
|Radial Expansion (mm)||1.205|
|Hardness (Shore D)||89|
|% Transmission (@600 nm)||>95|
|Standard size||Diameter - 12.70 mm, Thickness - 5.0mm|
|Standard Colors||Clear, High Blue, Low Blue, Low Green|
|UV Blocker||Available upon request|
|Precut||Depth - 1.5mm /2.0mm, Radius - 6.8mm|
|Note||Optional sizing and colors may be available on request.|
|We recommend maintaining a maximum Relative Humidity of 45% at 21±2°C|
|Base Curve||Speed (RPM)||Feed (mm/min)||Depth (mm)|
|Front Curve||Speed (RPM)||Feed (mm/min)||Depth (mm)|
Machining parameters and tool specifications are important for optimum surface finish and reaching precise target every time. Benz R&D validates the expansion values, target hitting ability and surface quality of every lot of Benz materials when machined strictly according to recommendations.
To polish use XPAL, Alox 721 or comparable material. Polishing should require a maximum of 20 seconds under the above machining conditions.
Place lenses into the Borate Buffer pH 7.2 isotonic saline and allow the lens to sit in solution while stirring for a minimum of 12 hours. Hydrate for the minimum time specified at constant temperature (T = 20 ± 2°C). A minimum of 10 ml per lens should be used for hydration.
After hydration, lenses should be rinsed, autoclaved, and stored in the pH 7.2 buffered saline solution for packaging and sterilization.
Borate Buffer pH 7.2; 295 mOs
NaCl 8.01 grams • H3BO3 2.47 grams • Na2B4O7 . 10H2O 0.14 grams
The weights for the buffered saline formulas are based on a 1 Liter Volume solution. The borate solution shows excellent performance through the sterilization process (autoclaving) and leaves the lenses free of residue.